5500 NL (newlook)
5500 MD (modular)
+ Compact design
+ Innovative effective pulsing technology
+ Low electronic failure statistics
+ Spare parts availability
– High Pump Chamber damage rate
– High HVPS (charger) failure rate
– Calibration sensitivity
– Unconscious, uninformed service abundance
Our company has serviced and repaired for the above-mentioned models are given.
Pump Chamber renovation – change
Simmer PS repair
Fiber Cable Repair
Moreover, the following parts we can provide.
IMPORTANT NOTICE 1
Pump Chamber renovation – as a result of changes in the Head 45 W for output that should take under 700 volts. This value you should read from the service`s Power meter.
45 W and 700 Volt Pump Chamber cannot provide criteria is problematic. In this case, you experience the following problem.
Excessive voltage rise in a short time
Losing energy of the flash lamps in a short time
High temperature alarm
Not stable shoot with the 15 head
Leaking water from the pump chamber
IMPORTANT NOTICE 2
Should make the re-calibration of your system is compared with the old value after especially flash lamp, focus lens and parts changing. Otherwise, your system will work inefficient . This control provides you to determine whether the part is original or not.
IMPORTANT NOTICE 3
Your system should run efficiently and with proper calibration. Person or persons who receive the service, select the correct criteria, do not make your partner of earnings
DAMAGED PUMP CHAMBER &
REPAIRED PUMP CHAMBER
CHAMBER VE ROD